<!--
 * @Author: cgl
 * @Date: 2023-04-18 15:12:26
 * @LastEditors: cgl
 * @LastEditTime: 2023-09-04 15:23:28
 * @FilePath: \public-health-emergencies-fed:\新放射\新radiation-hygiene-fe\src\views\person\itemControl\item12.vue
 * @Description: 
 * 
 * Copyright (c) 2023 by cgl/微盛信息产业有限公司, All Rights Reserved. 
-->
<template>
	<div>
		<table border="1" cellspacing="0" cellpadding="0" align="center">
			<tr class="title">
				<td rowspan="2">序号</td>
				<td rowspan="2">检测项目</td>
				<td rowspan="2" colspan="2">检测方法及条件</td>
				<td rowspan="2" width="20%">验收检测判定标准</td>
				<td rowspan="2">状态检测判定标准</td>
				<td rowspan="1" colspan="2">稳定性检测</td>
			</tr>

			<tr class="title">
				<td>判定标准</td>
				<td>周期</td>
			</tr>

			<tr>
				<td >1</td>
				<td >IP暗噪声</td>
				<td colspan="2">——</td>
				<td  class="viewfile" @click="viewInTable('G.2')">查看</td>
				<td  class="viewfile" @click="viewInTable('G.2')">查看</td>
				<td  class="viewfile" @click="viewInTable('G.2')">查看</td>
				<td >一个月</td>
			</tr>
			<tr>
				<td >2</td>
				<td >IP响应线性</td>
				<td colspan="2">单板</td>
				<td >R<sup>2</sup> ＞0.95</td>
				<td >——</td>
				<td >R<sup>2</sup> ＞0.95</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td >3</td>
				<td >IP响应均匀性</td>
				<td colspan="2">单板</td>
				<td >±10.0%内</td>
				<td >±10.0%内</td>
				<td >±10.0%内</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td >4</td>
				<td >IP响应一致性</td>
				<td colspan="2">多板</td>
				<td  class="viewfile" @click="viewInTable('G.3')">查看</td>
				<td  class="viewfile" @click="viewInTable('G.3')">查看</td>
				<td  class="viewfile" @click="viewInTable('G.3')">查看</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td >5</td>
				<td >IP 擦除完全性</td>
				<td colspan="2">4cmPMMA，0.1mmAl</td>
				<td >≤0.3</td>
				<td >——</td>
				<td >≤0.3</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td >6</td>
				<td >伪影</td>
				<td colspan="2">4cmPMMA</td>
				<td >无影响临床的伪影</td>
				<td >无影响临床的伪影</td>
				<td >无影响临床的伪影</td>
				<td >六个月</td>
			</tr>
			<tr>
				<td >7</td>
				<td >高对比度分辨力</td>
				<td colspan="2">高对比测试卡，水平和垂直放置</td>
				<td >≥90.0%厂家规定值，或≥70.0%（$f_{Nyquist}$），建立基线值</td>
				<td >≥90% <sup>a</sup></td>
				<td >——</td>
				<td >——</td>
			</tr>
			<tr>
				<td rowspan="6">8</td>
				<td rowspan="6">低对比度细节</td>
				<td rowspan="6" width="3%">按模体说明书选择曝光条件</td>
				<td >细节直径D <br> mm</td>
				<td >对比度</td>
				<td >对比度</td>
				<td rowspan="6">——</td>
				<td rowspan="6">——</td>
			</tr>
			<tr>
				<td >0.10≤D＜0.25</td>
				<td >＜23.0%</td>
				<td >＜23.0%</td>
			</tr>
			<tr>
				<td >0.25≤D＜0.5</td>
				<td >＜5.45%</td>
				<td >＜5.45%</td>
			</tr>
			<tr>
				<td >0.5≤D＜1.0</td>
				<td >＜2.35%</td>
				<td >＜2.35%</td>
			</tr>
			<tr>
				<td >1.0≤D＜2.0</td>
				<td >＜1.40%</td>
				<td >＜1.40%</td>
			</tr>
			<tr>
				<td >D≥2.0</td>
				<td >＜1.05%</td>
				<td >＜1.05%</td>
			</tr>
			<tr>
				<td colspan="8">
					<div style="text-align: left; margin-left: 20px;">
						<sup>a</sup> 与基线值比较。
					</div>
				</td>
			</tr>

		</table>

		<el-dialog
			title="不同生产厂家乳腺 CR 系统的暗噪声技术要求"
			:visible.sync="inTableVisible2"
			:close-on-click-modal="false"
			class="standarddialog"
      append-to-body
			v-dialogDrag="true"
		>
			<table class="custable" border="1" cellspacing="0" cellpadding="0" align="center">
				<tr>
					<td width="30%">厂家</td>
					<td width="40%">DDI</td>
					<td width="30%">暗噪声</td>
				</tr>
				<tr>
					<td>富士<br>(FUJI)</td>
					<td>S</td>
					<td>200</td>
				</tr>
				<tr>
					<td>爱克发<br>(Agfa)</td>
					<td>SAL<br>SALlog<br>PVIlog</td>
					<td>SAL≤134<br>SALlog15≤1000<br>PVIlog15≤1000<br>PVIlog16≤9503</td>
				</tr>
				<tr>
					<td>锐珂<br>(Carestream)</td>
					<td>EI</td>
					<td>≤150</td>
				</tr>
				<tr>
					<td>柯尼卡-美能达<br>(Konica-Minolta)</td>
					<td>S</td>
					<td>5000</td>
				</tr>
			</table>
			<div slot="footer">
				<el-button @click="inTableVisible2 = false">关 闭</el-button>
			</div>
		</el-dialog>

		<el-dialog
			title="不同生产厂家乳腺CR系统的IP响应一致性评价"
			:visible.sync="inTableVisible3"
			:close-on-click-modal="false"
			class="standarddialog"
      append-to-body
			v-dialogDrag="true"
		>
			<table class="custable" border="1" cellspacing="0" cellpadding="0" align="center">
				<tr>
					<td>检测条件</td>
					<td>富士、柯尼卡</td>
					<td>锐珂</td>
					<td>爱克发</td>
				</tr>
				<tr>
					<td>同样尺寸/型号多块 IP</td>
					<td>平均值±5.0%内</td>
					<td>平均值±20.0 内</td>
					<td>
						SAL 平均值±5.0%内<br>
						SALlog：平均值±200.0 内<br>
						PVIlog：平均值±290.0 内
					</td>
				</tr>
			</table>
			<div slot="footer">
				<el-button @click="inTableVisible3 = false">关 闭</el-button>
			</div>
		</el-dialog>
	</div>
	
</template>

<script>
export default {
  name: "item40",
  data() {
    return {
			inTableVisible2: false,
			inTableVisible3: false,
    };
  },
  created() {
  },
  methods: {
		viewInTable(type) {
			if(type == "G.2") {
				this.inTableVisible2 = true;
			}
			if(type == "G.3") {
				this.inTableVisible3 = true;
			}
		}
  }
};
</script>

<style lang="scss" scoped>
@import './content.scss';
</style>